Download ThinkRF S240v5 v5.1.3 installer
View Release Notes in the PDF form
This document describes the first maintenance release of the S240v5 Real-Time Spectrum Analyzer
(RTSA) software, version 5.1.3. It is concurrently released with PyRF3 API v3.0.7 (PyRF3 is a
Python 3 release of ThinkRF’s original Python 2 version of the PyRF).
This software is distributed as a single Microsoft Windows binary file
Reminder: S240v5 supports current RTSA devices such as R5500, R5550, R5700, & R5750, but not
the obsoleted WSA products such as the WSA5000. End of life products and the RFE ZIF mode are
still supported by the S240 application v4.3.0 or earlier.
2 Fixed Defects
Version 5.1.3 fixed the time-domain display when Span is 40MHz or lower.
Version 5.1.3 addresses the spur mitigation feature which does not work properly in previous
versions. The Spur Mitigation feature will now be available only when:
- The Stop frequency is not at maximum and the Start frequency is from 50MHz or higher; or
- The Stop frequency is at max and the Start frequency is from 52MHz or higher.
These constraints are because there is no frequency tuning in DD range (DC to 50MHz) & a shift in
the center frequency is needed for the spur mitigation algorithm to work.
Also, “Known Limitations of the Spur Migitation Feature” has been added to the S240v5 User Guide,
under “Configuring Spur Mitigation” section.
3 New Features
4 Other Changes
The following changes appear in Release version 5.1.2.
- Incorporated PyRF3 updates from v3.0.5 to v3.0.7.
- Updated the S240v5 User Guide to reflect the new information and limitations in “Configuring
Spur Mitigation” section.
5 Known Issues and Limitations
The following known issues and limitations exist in this version.
5.1 Demodulation Feature
The Demodulation feature is still in the beta stage; thus, there are known issues associated with it as
listed in this section.
5.1.1 Scrubbing and Stepping
Scrubbing and the associated ability to step through the signals has not been implemented in this
release. Although it is possible to see the location of the playback within a saved stream using the
slider below the graph section of the user interface, it is not yet possible to move back and forth
manually. Nor is it yet possible to move frame by frame in either direction.
5.1.2 Audio Record and Playback
The audio streams after demodulation cannot yet be saved and played independently of the IQ data.
To hear the audio again, it must be demodulated again from the original saved VRT stream.
5.1.3 AM Broadcast
The application cannot at this time tune down into the AM broadcast frequencies. This limitation may
be addressed in the future.
- If Span is set to under 40 MHz, then after switching back from running the Demodulation
feature for awhile, the data capture displayed in the main S240v5 panel might get stuck in
the ‘block’ capture mode. This means changing of the Center Frequency will work but the
sweep feature is no longer working. The work around for this is to use the Preset button or
reconnect to the device from Device > Connect.
- If Auto RBW is unselected, then after switching back from running the Demodulation feature
for awhile, the main S240v5’s Span or RBW will not be switched back properly. Suggest to
save the settings (File > Save Settings) before using the Demodulation feature so that the
settings could be reloaded from the saved *.config file (File > Load Settings).
5.2 Other Known Issues
- When Span is 25 MHz or lower, the data returned is of I & Q mode due to decimation used,
the DC offset correction is then activated. This means that any frequency input from a signal
generator that is aligned with the S240v5’s Center frequency will resulted in a ‘clipped’ input
signal. Recommend to have the input frequency of the signal generator and the S240v5’s
Center frequency to be at least 10 Hz off from each other to avoid such signal clipping.
- There are rare cases where the automated mode switching can result in a frozen display,
click on Pause Capture, Single Capture, then Continuous Capture to resume the capture.
The cause of this issue is not yet understood.
- Sometimes, when a Level Trigger is active and the plot is frozen by lack of triggered data,
switching to Free Run does not immediately restart continuous capture. This is easily
remedied by clicking Pause Capture and then Continuous Capture.
- Before running the installation, recommend turning off the anti-virus software. Some antivirus
software might set off a false positive toward the PyInstaller application, which is used
to generate the RTSA GUI executable from Python.
- During the installation, a Microsoft Certification warning might show-up. Accept the warning
to continue the installation.
- Fine Tuning is still not yet available.
- See “Known Limitations of the Spur Migitation Feature” in the S240v5 User Guide, under
“Configuring Spur Mitigation” section.
Download ThinkRF S240v5 v5.1.3 installer