Enter your search term here...
New support ticket
Check ticket status
+1 613 369 5104
How do I use S240 to connect to a ThinkRF R5500 Spectrum Analyzer?
There is no Ethernet port on my PC, can I use USB to connect to ThinkRF Spectrum Analyzer?
What is the warranty on ThinkRF Product?
Does ThinkRF provide extend warranty?
How often do I need to calibrate my ThinkRF box?
Does ThinkRF provide NIST-traceable calibration certificate?
What is the difference among Streaming, Sweeping and Scanning?
See all 7 articles
How do I contact ThinkRF?
D2030 27-30GHz Downconverter
ThinkRF D2030 Downconverter Datasheet
ThinkRF D2030 downconverter User Guide
ThinkRF D2030 Downconverter Programmer's Guide
Can I use a battery pack or a portable battery to power the D2030 Down Converter?
Is the power input always needed or can the D2030 Down Converter work passively?
Can I supply 10MHz reference without first powering up the D2030 device?
Do I need to reset D2030 when switching 10MHz reference clock?
What is the operational voltage range for D2030?
See all 9 articles
R5700 Real-Time Spectrum Analyzer with GNSS
ThinkRF R5700 Real-Time Spectrum Analyzer with Global Navigation Satellite System (GNSS) Datasheet
ThinkRF R5700 User Guide
ThinkRF R5700 Programmer's Guide
ThinkRF R5700 Firmware 1.0.1
R5500 Real Time Spectrum Analyzer
Updating the R5500's Firmware
Installing the USB drivers on your PC
Configure the R5500 analyzer's IP address through USB console
ThinkRF R5500 Datasheet
ThinkRF R5500 User Guide
ThinkRF R5500 Programmer's Guide
ThinkRF R5500 AppNote Using R5500 With HIF Output
ThinkRF R5500 AppNote Gain/Attenuation Settings for R5500-418 and -427
See all 13 articles
ThinkRF R5500 firmware v1.5.0
I cannot find com port when connecting to R5500 via USB console
Can the receiver handle 0 dBm input power level when receiver is not powered on?
Can I do my own calibration on the ThinkRF R5500 Spectrum Analyzer?
What is the required calibration interval?
What is the frequency accuracy on ThinkRF R5500 Spectrum Analyzer?
See all 10 articles
E300 Enabler ThinkRF - Keysight Vector Signal Analyzer
My R5500 device is on the network and discoverable, but I cannot connect to it from VSA
How do I purchase Keysight’s 89600 VSA software?
1. Installing the Keysight VSA 89600 and Keysight Connection Expert Software
2. Installing the ThinkRF E300 Enabler
3. Discovering ThinkRF R5500 Devices
4. Configuring the ThinkRF E300 Enabler
5. Viewing Device Information in Keysight VSA
See all 6 articles
ThinkRF E300 Enabler Product Brochure / Datasheet
P120 Vehicular Power Conditioner
ThinkRF P120 Vehicular Power Conditioner(VPC) Datasheet
ThinkRF P120 Vehicular Power Conditioner User Guide 1.1
ThinkRF S240 Product Brochure / Datasheet
ThinkRF S240 RTSA User Guide
When recording a spectrum of interest, are you recording video or recording / storing actual data?
Is the S240 software able to scan across a wide spectrum and then lock on to a signal of interest?
Is the S240 software free?
Does S240 RTSA software work on both R5500 and WSA5000
What is Spectrogram display in S240?
See all 9 articles
Saving/Recording Signal Data
Playing Back a Recorded Signal (Load data)
Is there a LabView API available?
ThinkRF RTSA LabVIEW API v2.0.0
ThinkRF RTSA AppNote Using the ThinkRF Real-Time Spectrum Analyzer with LabVIEW
C/C++ API 3.6.0
Using the ThinkRF R5500 Real-Time Spectrum Analyzer with C++
WSA5000 (Discontinued in May 2017 / end of support on May 31 2018)
ThinkRF WSA5000 Quick Start Guide
ThinkRF WSA5000 DataSheet
ThinkRF WSA5000 User Guide
ThinkRF WSA5000 Programmer's Guide
WSA5000 AppNote - Synchronized Sweep with IQout
See all 7 articles
WSA5000 Firmware 4.5.6
What is latest firmware for ThinkRF WSA5000 spectrum analyzer?