As of July 2020, we only have SCPI available for D4000. We are planning to provide Python API support for D4000 in the near future. Users can alway...
Fri, Jul 10, 2020 at 6:12 PM
There are 8 pre-select filters in total. they are: Filter 1: 38 - 41 GHz Filter 2: 36 - 39 GHz Filter 3: 34- 37 GHz Filter 4: 32 - 35 GHz Filt...
Fri, Sep 18, 2020 at 12:29 PM
That's for connecting it to other future external modules and have an external unit to control it.
Thu, Jul 2, 2020 at 2:58 PM
The front-end attenuator or user control attenuator range is from 0 dB to 31.5 dB with 0.5 dB step size.
Fri, Sep 18, 2020 at 12:35 PM
No. The IF OUT of D4000 has always spectrum non-inverted.
Thu, Dec 10, 2020 at 4:58 PM
The default IP setting for D4000 is on the Static IP address. It can be switched to a Dynamic IP address however D4000's firmware doesn't support de...
Tue, Sep 14, 2021 at 10:53 AM
This is due to a known issue where a PLL within the downconverter takes about 20 ms to change bands. Due to the frequency plan used in the unit, this occurs...
Fri, Feb 4, 2022 at 4:28 PM
Tue, Jan 10, 2023 at 4:08 PM
The input of the D4000 is AC couples only and it can not be configured. The coupling SCPI command is for querying only.
Tue, Jan 10, 2023 at 4:14 PM