Enter your search term here...
New support ticket
Check ticket status
+1 613 369 5104
R5500/R5550 Real Time Spectrum Analyzer
Updating the R5500's Firmware
Installing the USB drivers on your PC
Configure the R5500 analyzer's IP address through USB console
ThinkRF R5500 Datasheet
ThinkRF R5500 User Guide
ThinkRF R5500/R5550 Programmer's Guide
ThinkRF R5500 AppNote Using R5500 With HIF Output
ThinkRF R5500 AppNote Gain/Attenuation Settings for R5500-418 and -427
See all 14 articles
ThinkRF R5500/R5550 firmware 1.6.3
ThinkRF R5500/R5550 Firmware v1.6.2
ThinkRF R55xx Firmware 1.6.1
ThinkRF R5500 Firmware 1.6.0
ThinkRF R5500 firmware v1.5.0
I cannot find com port when connecting to R5500 via USB console
Can the receiver handle 0 dBm input power level when receiver is not powered on?
Can I do my own calibration on the ThinkRF R5500 Spectrum Analyzer?
What is the required calibration interval?
I am considering upgrading to a ThinkRF R5500 product. Do you have any information comparing the ThinkRF R5500 and WSA5000 products?
See all 17 articles