Enter your search term here...
New support ticket
Check ticket status
+1 613 369 5104
R5500/R5550 Real Time Spectrum Analyzer
Updating the R55x0's Firmware
Installing the USB drivers on your PC
Configure the R55x0's static IP address through its USB console
Set up a ThinkRF device remotely
ThinkRF R5500/R5550 Datasheet
ThinkRF R5500/R5550 User Guide
ThinkRF R5500/R5550 Programmer's Guide
AppNote: Gain/Attenuation Settings for ThinkRF R5xx0-418 and -427
AppNote: Performing Real Time Data Acquisition With ThinkRF Real-Time Spectrum Analyzer R5xx0
See all 14 articles
I cannot find com port when connecting to R5500 via USB console
What is the sweep rate of R5500 Spectrum Analyzer?
Can the receiver handle 0 dBm input power level when receiver is not powered on?
Do I have to warm up ThinkRF Spectrum Analyzer for 30 mins before each use?
What is ZIF, SH, SHN, HDR, or DD mode for R5xx0 products and when to use which?
See all 31 articles
What PLL source used in factory calibration?
What decimation setting is used during factory calibration?
Is the reference input signal a CW?
What input power level is used during the factory calibration?
Is CW signal placed at center during factory calibration?
See all 6 articles
AppNote: Using ThinkRF R55x0 With HIF Output
How do I use HIF function on ThinkRF R5500 Spectrum Analyzer?如何使用R5500的HIF功能？
Is R5500/R5550 HIF output calibrated?
Is it normal to see signal coming out of the HIF port?
Why ThinkRF does not offer a WBIQ option with internal digitizer?
Is IQ output calibrated?
How is the internal digitiser outputs made available to the user for WBIQ units?
Does DD mode support WBIQ?