Micro D type cable can be used for GPIO port, an example can be found on Digi-Key. Part number: CCA-025-I36R152
Fri, Sep 21, 2018 at 4:34 PM
DANL for 408P is the same as for 418 but up to 8GHz. 408P uses the same daughter card as 418.
Tue, Jun 11, 2019 at 2:58 PM
No, sampling speed could not be adjusted in R5500. When users decimate, the sampling speed is effectively reduced.
Tue, Oct 1, 2019 at 5:03 PM
Is there any sort of report generated after scanning across frequencies besides seeing spectrum on the user interface?
Spectrum monitoring report generated via UI or API as csv file including: time, frequency, power level, bandwidth, (any other measurements).
Mon, Jan 6, 2020 at 4:00 PM
ThinkRF does not support Automatic Gain Control at this moment.
Tue, Oct 1, 2019 at 5:01 PM
RTSA7550/R5500 does not support phase measurement, such as phase shift, phase noise.
Fri, Jun 14, 2019 at 5:17 PM
IBW can be set to 100MHz (ZIF mode), 40MHz (SH mode), 10MHz (SHN mode), 100kHz (HDR mode). It can be also set to other values using Decimation for ZIF, SH, ...
Wed, Oct 16, 2019 at 2:48 PM
R5500-408E wight 3 lbs, or 1360.5g.
Thu, Jul 2, 2020 at 12:09 PM
For R5500/R5550 and R5700/R5750, the tuning resolution (or min frequency step size) is 10Hz. Detail can be found at "[:SENSe]:FREQuency:CENTer&quo...
Tue, Aug 18, 2020 at 1:17 PM
The noise figure of the entire tuner (including the digitizer) is reflected in the DANL of our product. Please refer to the ThinkRF R5500/R5550 Datasheet. T...
Fri, Aug 14, 2020 at 4:06 PM