Micro D type cable can be used for GPIO port, an example can be found on Digi-Key. Part number: CCA-025-I36R152
Fri, Sep 21, 2018 at 4:34 PM
This is a English/Chinese languages version. This FAQ will walk you through how to use ThinkRF R5500 HIF function. 此文将简单阐述如何使用ThinkRF R5500的HIF功能 Run...
Tue, Dec 4, 2018 at 1:34 PM
DANL for 408P is the same as for 418 but up to 8GHz. 408P uses the same daughter card as 418.
Tue, Jun 11, 2019 at 2:58 PM
No, sampling speed could not be adjusted in R5500. When users decimate, the sampling speed is effectively reduced.
Tue, Oct 1, 2019 at 5:03 PM
Is there any sort of report generated after scanning across frequencies besides seeing spectrum on the user interface?
Spectrum monitoring report generated via UI or API as csv file including: time, frequency, power level, bandwidth, (any other measurements).
Mon, Jan 6, 2020 at 4:00 PM
ThinkRF does not support Automatic Gain Control at this moment.
Tue, Oct 1, 2019 at 5:01 PM
RTSA7550/R5500 does not support phase measurement, such as phase shift, phase noise.
Fri, Jun 14, 2019 at 5:17 PM
For the upper receiver section in the image below. When IQ Out is selected, the switch goes to IQ Out. Data no longer goes to the internal Digitizer (lowe...
Tue, Oct 1, 2019 at 4:59 PM
IBW can be set to 100MHz (ZIF mode), 40MHz (SH mode), 10MHz (SHN mode), 100kHz (HDR mode). It can be also set to other values using Decimation for ZIF, SH, ...
Wed, Oct 16, 2019 at 2:48 PM